This document specifies the characteristics of, and corresponding test methods for, wrought stainless steel containing a mass fraction of 0,25 % to 0,50 % nitrogen for use in the manufacture of surgical implants for which high levels of strength and corrosion resistance are required.
NOTE 1 The mechanical properties of a sample obtained from a finished product made of this alloy can differ from those specified in this document.
NOTE 2 Requirements for other stainless steels for implants for surgery can be found in ISO 5832-1.
Status: PublishedPublication date: 2019-02
Edition: 3Number of pages: 5
Technical Committee: ISO/TC 150/SC 1 Materials
- ICS :
- 11.040.40 Implants for surgery, prosthetics and orthotics
This standard contributes to the following Sustainable Development Goal:
Buy this standard
|std 1 40||PDF + ePub|
|std 2 40||Paper|
ISO 5832-9:2019Stage: 60.60
Got a question?
Check out our FAQs
Monday to Friday - 09:00-12:00, 14:00-17:00 (UTC+1)
Keep up to date with ISO
Sign up to our newsletter for the latest news, views and product information.